Sidmar Hydromassage Tables for Professional Use
The Sidmar Massage Time Pro hydro therapy massage table is used to treat soft tissue, muscle, joint, and bone injuries. It helps to reduce aches and pains by increasing blood circulation. The Massage Time Pro table can be used in multiple healthcare settings, such as in hospitals, physical therapy, chiropractic, and medical clinics. Used by more than 6,000 clinicians every day, the Massage Time Pro table specifically massages the neck, back and shoulders and can massage the legs when the patient turns the opposite way on the table.
Need Service? If you're located within 30 miles of Philadelphia, PA we're happy to service your Sidmar unit at your office. We bill hourly for labor ($125/hr). Call us if you need,
- New system installation.
- Jet system replacement.
- Pump replacement.
- PVC pipe repairs.
- Installation of other parts such as relays, timers, flow controls, and heaters.
- General maintenance.
- Feels like a Jacuzzi, without undressing or getting wet.
- Very quiet and is soothing for patients.
- Requires minimal maintenance except to add distilled water and Clar-Lub conditioner every three months to top off the water level.
- Individual adjustments to control heat and water pressure, so it is easier to accommodate all patients.
- Use pre- or post-adjustment to relax muscles.
- Use before or after decompression therapy to help relax the muscles and to reduce pain.
- Enhances exercise and strength building programs.
- Available in multiple colors such as Black, Blue Ridge, Teal, Forest Green, Tan, Burgundy, Greystone, and Navy Blue.
- Easily cleaned with mild soap on a damp cloth.
To ensure smooth and long-term operation of the Massage Time Pro hydromassage bed.
- Upon installation, set the table edge at least 6 to 8 inches away from any wall.
- Please alert patients to remove any sharp objects from their pockets. Including pens, keys, and cell phones.
- Avoid moving or lifting the hydromassage table when it is full of water because this can stress the frame and pipe joints. We suggest that the water be drained and stored prior to moving the unit significant distances.
- Check the PVC fittings at least once per month to ensure that they are hand-tight. Do not use a wrench or other device to tighten these fittings as it may stress or crack the joint and cause a leak.
- Check the water level and add distilled water with a bottle of Clar-Lub conditioner at least once every three months.
Sidmar Pro S10 Hydromassage Table Checklist:
The following is my list of
things to consider before/during/after installation of your Pro S10
hydromassage table. This comes from years of hydromassage table installs and repairs. - Anthony Engel
- Power outlet: If possible, try to
dedicate a circuit to the Pro S10, i.e. avoid plugging in on same circuit as
refrigerator, AC, dehumidifier, etc.
- I recommend positioning the long side
of your Pro S10 along a wall and then installing a grab bar(s) along the same
wall so patients can self-assist with getting on and off the unit. It takes
some pressure off the neck, lower back, and shoulders.
- Pallet: While I have handled these on
my own using a hand-truck, two people are way better than one for moving this
- Pallet: Follow Sidmarís directions for
unpacking from box, i.e. lift the unit up and out. Avoid attacking the box with
a sharp blade as you want to avoid cutting the Pro S10.
- Once out of the box, follow Sidmarís
instructions for set up. Do NOT plug in the Pro S10 yet.
- Remove metal wristwatch, bracelets,
and any other objects that may scratch or puncture the mattress. The material
is tough, but I try to be careful. I typically
- I typically remove the three side
panels (2 short ones by controls and 1 long one on opposite side) so I can see
the inner workings on the bottom.
- Ensure that the heater is in the
general vicinity of the black outline.
- Ensure that the mattress is unfurled
and that the protective outer skirt is completely spread out as if it is
holding the mattress in a box. The outer skirt is your insurance policy against
flooding should the mattress become punctured.
- Check all PVC fittings to ensure that
they are Hand-Tight (Do Not use a wrench to tighten because the force can crack
the fitting). For clinics, I recommend checking these fittings once per month.
For everyone else, once per quarter is fine.
- Continue with Sidmarís instructions to
fill the unit. I typically pour the Clar-Lub conditioner in first (so I donít
forget). The unit takes 21 gallons of distilled water.
- While filling, I typically stop (put stopper
and cap on mattress fill spout) and reach under the mattress from all sides 2
or 3 times to work out the kinks and to ensure that itís resting evenly on the
table. I also adjust the skirt so that it is not binding and that it is fully
encapsulating the inner mattress.
- Once the fill is complete, expel as
much air as possible from the mattress. This is a technique that gets better
- Replace stopper and cap on pour spout.
I typically will take a paper towel or clean terry rag and wipe away excess
water all around mattress including under it, between mattress and skirt (just
lift each corner of the mattress and wipe as best as possible). This makes it
easier to spot a mattress leak (highly unlikely but I like to be sure).
- Plug in the unit and youíre ready to test
Ė Without a human. I typically turn the pressure regulator to low and set the
timer for 20 minutes. Try not to handle the mattress while itís running because
you donít want the steel jets to scrape against the inner mattress wall. While
running, look above and below and, using a flashlight, look around to be sure
you donít see drips.
- When the unit turns off, lift the
corners of the mattress to check for water again. Remember that there could be
some residual water that spilled during the fill. Just take your time to ensure
that things are dry.
- Now test With human. First: Remove
sharp objects from all pockets and belts. The unit should be Off when lying
down though if youíre alone then itís okay to turn it on first. When mounting
unit, it is critical that your tail bone is on the hard surface. This will
ensure that direct weight is not being placed on the steel jet. While sitting
on the hard surface, pivot slowly while lying back slowly until your back is
stretched evenly across the mattress and your feet are resting on the footrest.
- For Clinics: if you want to guard
against leaks, it is important to guide your patients on the proper way to sit
and then recline onto the mattress. Too many times I have had to repair or
replace a mattress because a patient placed their tailbone directly onto the
jet instead of the hard surface. These unfortunate events are avoidable when staff
takes a little time to educate and demonstrate proper technique. Remember to
ask patients to remove sharp objects from all pockets and belts.